Rhythmedics® Bio-Pulses™ Interrupt Harmful ElectroMagnetic Radiation (EMR)
Active versus Passive EMR Interruption
Rhythmedics instruments are active devices emitting Fibonacci based Bio-Pulses to actively interrupt harmful EMR. In contrast, Q-Link’s pendant, which has been touted for its ability to transform harmful EMR into positive energy, is a passive device. We acquired a new SRT-3 Q-Link and using our state-of-the-art laboratory conducted a series of tests comparing the measurable and viewable affects of both the Q-Link and Rhythmedics Bio-Pulses to interrupt EMR from three common household electrical products; A) 5.8 GHz Cordless Phone, B) 12 volt Plug-in Wall Transformer used to recharge electronic devices, and C) desktop fluorescent lamp’s power supply.
Three images are displayed for each electrical product tested. The first image shows the raw EMR source waveforms, the second shows the EMR waveform with the Q-Link between the probe and source, and the third shows the EMR waveform with the Rhythmedics Bio-Pulse instrument between the probe and source. In every case the Q-Link shows no affect on the EMR waveform, whereas the Rhythmedics Instrument clearly interrupts the EMR.
1. CORDLESS PHONE EMR (First Image No Protection, Second Image Q-Link and Third Image Rhythmedics Bio-Pulses)
2. WALL TRANSFORMER EMR (First Image No Protection, Second Image Q-Link and Third Image Rhythmedics Bio-Pulses)
3. FLUORESCENT LAMP POWER SUPPLY EMR (First Image No Protection, Second Image Q-Link and Third Image Rhythmedics Bio-Pulses)
Notes: All measurements were made with a Tektronix TDS 3034 Digital Phosphor Oscilloscope. A 10 milliHenry ferrite core inductor with a 10,000 ohm termination resistor was used as the probe. Measurements were made at a distance of 1 inch from the EMR source to eliminate ambient EMR sources. Once a raw EMR signal was captured on the digital oscilloscope the SRT-3 Q-Link was inserted between the EMR source and the probe and the second image was captured. Then SRT-3 was removed and the Rhythmedics Bio-Pulse Instrument was placed between the EMR source and probe and the third image was captured. All settings for the Oscilloscope are displayed on its screen.
The images above clearly show a passive device makes no observable changes to the EMR waveform from common household sources. Whereas the images clearly show the Rhythmedics Bio-Pulses interrupts the continuous EMR. This is critical as studies of EMR have shown it is necessary to interrupt the continuity or coherence of the EMR waveform in order to prevent it from adversely affecting the health of humans and animals. All Rhythmedics Instruments create Bio-Pulses that effectively interrupts harmful EMR as shown above.